The MS236WT-A detects up to 10 touches simultaneously and permits smooth and accurate stylus input. It ignores erroneous input, for example when the ball of the hand touches the surface.
Precise, high-contrast, bright and crisp screen
Thanks to the high 2 Megapixels (colour) resolution, a strong contrast ratio of 1000:1 and stable brightness of up to 260 cd/m2, the monitor offers excellent image quality. Even the differences between the finest details are shown – regardless of your viewing angle. This is a great advantage if multiple physicians are looking at the screen.
An internal 10 bit LUT controls color rendering. The color gradations are considerably more exact and finer than the standard 8 bit LUTs, making clinical review easier.
Operation with finger or touch pen
The MS236WT-A contains reliable capacitive touch technology (PCAP) that detects movements with very high precision. Hence you can easily capture even small writing and character input via the screen.
The display extends right up to the bezels – this means that the entire touch surface can be used without disruptions right up to the edges, which is particularly convenient for swiping and scrolling motions.
The monitor can detect up to ten simultaneous touches, which means that multiple individuals can operate the monitor at the same time. Furthermore, the touchscreen is equipped with hand ball recognition. While writing, users can place their hands or arms on the monitor without accidentally triggering an input.
With a hardness of 5H, the glass surface is insensitive to scratches and guarantees to recognize up to 50 million touches.
The monitor is equipped with technology that calibrates the touch sensitivity to minimise incorrect touch response due to environmental factors and electromagnetic noise. This ensures that the screen maintains accurate touch interaction during use.
Users can easily turn touch detection on or off by pushing the button conveniently located on the monitor's outer rim. A button is located on two sides of the monitor for easy access.
The multitouch interface is supported by standard Windows 11, 10 and 8.1 drivers. You can operate the touch panel without additional drivers by simply connecting the USB cable.
In addition to Windows standard touch drivers, a dedicated touch driver is also available to support touch interaction sounds and mouse emulation.
Efficiency in diagnostics
EIZO measures and adjusts each tone of grey carefully so that the monitors comply with the DICOM® standard when delivered from the factory. The result is a particularly consistent gradation of grey tones, allowing for optimal radiological clinical reviews.
One DisplayPort and one HDMI input allow the connection of two digital signals simultaneously. A D-sub input is available for an analog signal. You can connect up to three computers simultaneously. You can switch between them automatically or manually, if desired.
A loudspeaker and headphone connection are subtly integrated in the housing. The audio connection to the monitor takes place via the DisplayPort or HDMI connection or the audio jack.
The equipped stand provides flexible positioning to allow the monitor to rest on your desk like a laptop and recline back to your desired angle. This makes it easy to place the monitor in the ideal position for comfortable use in any environment.
Environmentally and socially conscious production
The MS236WT-A is produced in a socially responsible way. It is free of child labour and forced labour. Suppliers along the supply chain have been carefully selected and they have also committed themselves to produce in a socially responsible way. This applies in particular to conflict minerals. We present a detailed report about our social responsibility annually and voluntarily.
Each MS236WT-A is manufactured in our own factory, which implements an environmental management system in accordance with ISO 14001. This includes measures to reduce waste, wastewater and emissions, resource and energy consumption, as well as to encourage environmentally conscious behaviour among employees. We publicly report on these measures on an annual basis.
The MS236WT-A is designed to have a long service life and normally outlasts the warranty period by some distance. Replacement parts are available many years after production has ceased. The entire lifecycle takes into account the impact on the environment as the longevity of the product and the fact it can be repaired saves resources and protects the environment. When designing the MS236WT-A, we took a minimalistic approach to our resources by using high-quality components and materials, as well as a careful production process.
Highest investment security
EIZO grants a three-year warranty. This is possible thanks to the highly developed production process based on a simple principle of success: sophisticated and innovative technology, made from high-end materials.
Graphics board recommendation
For precise diagnostics
The EIZO graphics card supports the properties, functions, and settings of the RadiForce MS236WT-A optimally. It enables precise diagnosis and can control several monitors simultaneously. EIZO offers technical support and warranty service for the graphics card.
Is there something else you need?
|Areas of application||Healthcare|
|Screen size [in inches]||23|
|Screen size [in cm]||58|
|Viewable image size (width x height) [in mm]||509,2 x 286,4|
|Resolution in MP||2 Megapixels (colour)|
|Ideal and recommended resolution||1920 x 1080 (Full HD)|
|Pixel pitch [in mm]||0,265 x 0,265|
|Supported resolutions||1920 x 1080 (Full HD), 1680 x 1050, 1280 x 1024, 1024 x 768, 800 x 600, 720 x 400, 640 x 480|
|Max. viewing angle horizontal||178|
|Max. viewing angle vertical||178|
|Number of colors or greyscale||16.7 million colors (D-Sub, 8 Bit), 16.7 million colors (DisplayPort, 8 Bit), 16.7 million colors (HDMI, 8 Bit)|
|Color palette/look-up table||1.06 billion color tones / 10 Bit|
|Max. color space (typical)||sRGB (100%)|
|Max. brightness (typical) [in cd/m²]||260|
|Max. dark room contrast (typical)||1000:1|
|Response time grey-grey alternation (typical) [in ms]||11|
|Max. refresh rate [in Hz]||60|
|Features & Operation|
|Preset color/greyscale modes||2x manual memory locations, sRGB, DICOM|
|DICOM tone curve|
|Automatic signal input recognition|
|On-screen menu languages||de, en, fr, es, it, se|
|Adjustment options||Color Mode, Brightness, Contrast, Color temperature/White point, Gamma, Overdrive, Colour tone, Color saturation, Resolution, Scaling, Touch function On/Off, OSD language, Signal input, Volume, Key lock|
|Touch panel type||Projected capacitive (PCAP)|
|Touch compatible OS||Windows 8.1 (64-bit/32-bit), Windows 11, Windows 10|
|Signal inputs||D-Sub, DisplayPort (HDCP 1.3), HDMI (HDCP 1.4)|
|USB specification||USB 5Gbps (USB 3)|
|USB upstream ports||1 x type B|
|USB downstream ports||2x type A|
|Graphic signal||RGB Analog, DisplayPort, HDMI (RGB, YUV)|
|Audio input||3,5 mm stereo jack, DisplayPort, HDMI|
|Frequency||Display Port: 31-68 kHz/59-61 Hz; HDMI: 31-68 kHz/59-61 Hz; D-Sub: 31-81 kHz/55-76 Hz|
|Power consumption (typical) [in watts]||15|
|Maximum Power Consumption [in watts]||47 (at maximum brightness with all signal inputs and USB ports in use)|
|Max. Power consumption in stand-by mode [in watts]||0.5|
|Power consumption with power switch off [in watts]||0|
|Energy consumption/1000h [in kWh]||17|
|Power supply||AC 100-240V, 50/60Hz|
|Dimensions & Weight|
|Dimensions (incl. stand) (width x height x depth) [in mm]||556,7 x 143,9-353,9 x 216,3-401,3|
|Weight (incl. stand) [in kg]||6.6|
|Dimensions (without stand) (width x height x depth) [in mm]||556,7 x 339,2 x 54|
|Weight (without stand) [in kg]||6|
|Dimension drawing (PDF)||Dimension drawing (PDF)|
|Tiltability||15 / 70|
|Hole spacing||100 x 100|
|Certification & Standards|
|Operating temperature||5 - 35 °C / 20 - 80 % (R.H., non condensing)|
|Certification||CE (Medical Device), CSA C22.2 Nr. 601-1, EN60601-1, IEC60601-1, RCM, FCC-B, CAN ICES-3 (B), PSE, VCCI-B, RoHS, WEEE, China RoHS, CCC, BIS, EAC|
|Software & Accessories|
|Other box contents||Touch-Pen inkl. Halterung, USB cable (Type A - Type B), Manual via download, Power cord|
|Accessories||PM200-K, CP200, PP100-K, TP5|
|Recommended graphics card||MED-XN51LP|
|Warranty periode||3 years|
Downloads & Service
Around your product
Test the properties and parametres of your monitor quickly and easily with software developed in-house at EIZO.
The EIZO software is capable of complete quality management – from calibration through asset management to acceptance and constancy testing.
RadiCS LE quality control software calibrates EIZO RadiForce monitors and manages the calibration data.
EIZO software for network-supported quality management in larger establishments – with remote function for monitors.
EIZO monitors generally have two or more signal inputs. You can easily switch between two or more computers by pressing a button on the front of the monitor. For devices with KVM switch functionality, the USB hub also switches to the respective computer at the same time.
This may be a problem with the graphics card driver. Please install the latest driver for your graphics card and restart the PC with the monitor.
Older graphics cards may not be able to display this resolution. In this case, you will require a new graphics card with the matching output.
Firstly, please check the resolution of the LCD screen. Check the recommended resolution stated for the respective LCD screen. If you have set a lower resolution than recommended, this may result in blurring.
The LCD display is comprised of a fixed number of image elements (pixels), which light up to display the contents of the screen. The image is comprised of horizontal and vertical pixels. LCDs achieve the best image quality if all pixels can be allocated data, i.e. if the resolution configured on the computer corresponds to that of the LCD display.
The color temperature is a benchmark for measuring the white tone and is stated in Kelvin. At high temperatures, the white tone appears slightly blue, while it has a more red tone at low temperatures.
5000 K: Often used in the printing industry.
6500 K: Suitable for displaying photos and video images.
The color tone rendering is relative to the change of the input signal level. This relationship is generally known as the gamma characteristic. In image reproduction, low gamma values result in white images and high gamma values result in high-contrast images.
International standard for an RGB color space. A color space has been defined, which most devices can comply with, to enable color matching between various applications and devices, such as monitors, scanners, and digital cameras. sRGB images, which are detected or issued on "sRBG devices" are limited to the full sRBG color range and no colors outside the sRBG color space. Color allocation, white point, and gamma are permanently defined.
In connection with a USB-compatible computer, the monitor acts as a hub, to which other USB-compatible peripherals can be connected.
The viewing angle relates to the angle, beneath which the contrast is better than 10:1. These are normally values of 160° to 178°. These values do not provide any information about what enormous differences in contrast may result within the viewing angle. The contrasts should not change significantly for users with a perpendicular view and a slightly side-on view. It must not make a difference whether the viewer is looking at the middle of the image or the edge of the image. How much the contrast stability varies with different LCD technologies is best checked using a measurement diagram or by a direct visual comparison. The more stable the contrast in the user's viewing frustum, the better the image rendering.
This is where the signal values are allocated to the color tone values for the output in the monitor. EIZO monitors are 24, 16, 14, or 10 bit instead of the conventional 8 bit look-up table (LUT) for precise color control. Precision work, performed by a high-performance chip developed in-house at EIZO.
Balanced brightness is crucial for brilliant color reproduction. Traditional LCD monitors need one to two hours to even out their brightness. However, even then, they are still sensitive to temperature deviations. Patented electronics regulate the backlight, depending on the model. They ensure constant brightness, regardless of the operating times and temperature.
This is a digital signal interface for images and sound signals for computer monitors with a high transmission bandwidth and a compact plug design.
Contrast is the ratio of white to black and for monitors is normally stated as a dark room contrast from a controlled laboratory environment. Typical contrasts on printed paper are 6:1 for newspapers, 31:1 for laser printers, 49:1 for photo paper. For example, dark room contrasts cannot be achieved in the office because an extremely deep black cannot be achieved in normal ambient light – even with the screen turned off.
Please contact our service partner for this:
Moosacherstrasse 6, Au
8820 Waedenswil ZH
Free service hotline +41 44 782 28 28. Response times are Monday to Friday from 08:00 - 12:00 and 13:00 - 17:00.
The letters behind the numbers have the following meanings:
The EIZO on-site exchange service is handled by our service partner ESAG. Please contact the hotline: +41 44 782 28 28. Response times are Monday to Friday from 08:00 - 12:00 and 13:00 - 17:00. If the replacement of your device is necessary, an equivalent replacement will be ready for you within 1-2 working days. A driver will come to you and deliver the replacement device to the goods receiving department or reception. He will take the defective device directly from there.
Screens have millions of pixels, which each have a surface area of less than a hundredth of a square millimeter. These elements may have defects, which are expressed by them being constantly lit or unlit. Visibility is fundamentally dependent on the surrounding area. For example, lit elements are almost invisible in a bright environment. Each LCD screen is unique in terms of pixel errors.
To give you a clear statement about what frequency of pixel errors EIZO deems tolerable, EIZO screen models are tested according to internationally recognized standards. On average the actual levels are in fact far below the tolerances across all of the devices. LCD modules that do not meet these requirements are not used at EIZO. Depending on the model, EIZO monitors have a zero pixel error guarantee for completely illuminated sub-pixels (partial pixels ISO 9241-307). Valid: six months from the purchase date.
Repairs and on-site replacement service are handled by our repair center:
Moosacherstrasse 6, Au
8820 Waedenswil ZH
Free service hotline +41 44 782 28 28. Response times are Monday to Friday from 08:00 - 12:00 and 13:00 - 17:00.
For repairs outside the warranty, you will receive a cost estimate immediately.
Since the German Electrical and Electronic Equipment Act (ElektroG) entered into force on March 24, 2006, old electrical equipment can no longer be disposed of with the normal waste. It must be disposed of in an environmentally friendly manner and recycled. This also applies to old EIZO devices.
If you are a private end consumer, you should simply take your old monitor to a local collection center. You will not be charged for this. You can find out where your nearest collection center is from your local authority.
If you are a business customer, simply contact us.
No, you do not need a driver. The monitors are identified via the signaling line and then installed as a standard or plug & play monitor.
Digital signal encryption system as copy protection for digital contents, for example for videos, music, etc. The transmission of digital contents is protected as the contents are encrypted on the transmitter's side, sent via a DVI output and once again decrypted on the recipient's side. Digital contents may not be reproduced if both the transmitter and recipient device do not support HDCP.
EIZO RadiForce monitors that can be used in close proximity to the patient fulfill the requirements of EN60601-1 or the CE-Medical Device Directive. The product data sheet describes whether a RadiForce model provides this level of electrical safety.
EIZO RadiForce LCD monitors can generally be removed from their bases. They are equipped with four VESA-compatible drill holes to attach other holding devices on the reverse. Before they are fitted, the maximum load-bearing capacity of the mounting device should be checked.
To keep your monitor looking like new at all times and to prolong the service life of your device, we recommend cleaning it regularly. Clean the casing and LCD surface as described below:
IMPORTANT: Never use thinner, benzene, alcohol (ethanol, methanol or isopropyl alcohol), scouring agents or other strong solvents as these may damage the casing and the LCD display.
Remove marks by wiping the casing with a soft cloth, which you have previously moistened with a mild cleaning agent. Never spray wax or cleaning agents directly on to the casing. (The manual contains exact details.)
Where necessary, the monitor surface may be cleaned using a soft cloth (for example a cotton cloth or glasses cloth). Stubborn marks can generally be removed by wetting the cloth with a bit of water to increase the cleaning strength.
Homogeneous luminance distribution and good color purity are key to achieving very accurate image rendering. The Digital Uniformity Equalizer (DUE) circuit takes care of this. It automatically corrects non-uniformities in terms of luminance and chrominance for all tone values across the entire image area – pixel by pixel.
The warranty for EIZO RadiForce monitors is 5 years for most models. EIZO recommends an image brightness that is considerably higher than the minimum requirements in accordance with QS RL for monitors used for medical diagnostics. This recommended brightness is guaranteed by EIZO.
The corresponding product data sheets show which guarantee applies to which RadiForce model.
Both the installation wizard in RadiCS 5 and the corresponding functionality in RadiNET Pro enable an upgrade including data migration.
An important prerequisite for this is that the installed RadiCS 4 Version must be Version 4.6.1 or higher. For that reason, you should carry out an update beforehand for older RadiCS 4 installations.
EIZO monitors support DisplayPort Alternate Mode with their USB-C interface. This means they can correctly display image signals from Thunderbolt 3 or Thunderbolt 4 ports via USB-C connection. The USB-C signal cable supplied by EIZO with the monitor has the optimal suitability.
Alternatively, monitors with a DisplayPort socket can be connected using a suitable cable. The CP200 signal cable (USB-C (TB3/TB4) to DisplayPort) provided by EIZO is the most suitable.
A connection from Thunderbolt 3 or Thunderbolt 4 ports to the HDMI interface of the monitor can work with suitable active adapters/cables. However, we recommend using the USB-C or DisplayPort interface of the monitor.
Preparing for the move:
First make sure that RadiCS can be opened on the previous computer. Our recommendation: If necessary, update RadiCS on the previous computer to the same version that you will use on the new computer.
Carry out the move:
EIZO has developed the RadiCS software specifically for quality assurance for image display devices in the RadiForce range. It offers extensive tests and automatic alignment to guarantee constant and consistent image reproduction on all RadiForce LCD monitors, but also supports screens from other manufacturers. The software automatically executes test images and helps to perform measurements and assessments in accordance with the guidelines in a cost-effective and time-effective manner. For example, measurement results are automatically transferred to the documentation and archived and can be centrally managed using the optional RadiNET Pro software. A calendar with a reminder function ensures timely checks are performed. A suitable measurement device is required to measure the luminance, including veiling luminance. Measured values from various luminance meters are automatically read by RadiCS.
According to the Federal Ministry for the Environment, Nature Conservation and Nuclear Safety (BMU), the quality assurance guidelines are according to the X-ray ordinance (QS-RL): "Quality assurance measures for X-ray devices to examine or treat humans".
DIN V 6868-157. This DIN governs the acceptance test and constancy test for image display systems in radiography in order to ensure the necessary image quality.
In accordance with DIN 6868-157, different minimum requirements are placed on image display devices depending on the body region / method. The EIZO RadiCS quality assurance software is divided into classes I to VIII. The following table shows a classification of the classes and minimum requirements in this regard.
Note: An acceptance test is necessary for each image display system for an evaluation in accordance with DIN 6868-157.
EIZO provides up to a five-year warranty, although some devices have a different warranty period. You can find out the exact warranty periods for each product either in the warranty terms and conditions above, in the datasheet, or on the EIZO international website. The long warranty periods are made possible by a highly developed manufacturing process based on a simple principle of success: sophisticated and innovative monitor technology, manufactured from high-end materials. For you, the 5-year warranty means a high level of investment security and savings on follow-up costs (total cost of ownership), which would be incurred without warranty service, for example, for maintenance and repairs.